Oct. 20, 2021
Dosing syringe is used for sterilization, generally using ethylene oxide sterilization, but if you do not pay attention to these points, it is easy to have hidden dangers or affect the sterilization effect.
1. The ethylene oxide sterilizer must be placed in a well-ventilated place. Do not place it near the fire source. In order to facilitate maintenance and regular maintenance, 51cm of space should be reserved on each side (including the top) of the ethylene oxide sterilizer. Special exhaust ducts should be installed and completely isolated from other exhaust ducts in the sterilization site.
2. Items that need to be sterilized must be thoroughly cleaned. Note that they cannot be cleaned with saline. There should be no water droplets or too much water on the sterilized items to avoid dilution and hydrolysis of ethylene oxide. If the packaging material is changed, it should be verified to ensure the reliability of the sterilization of the sterilized items.
3. There should be gaps on the top, bottom, left and right of the loaded items in the sterilization cabinet (the sterilized items cannot touch the cabinet wall), and the items should be placed in a metal mesh basket or on a metal net rack; the amount of items loaded should not exceed the total volume of the cabinet 80% of it.
4. Sterilization treatment: It should be carried out in accordance with the operating instructions of the manufacturer of the ethylene oxide sterilizer; select the appropriate sterilization parameters according to the type, packaging, loading volume and method of the plastic dosing syringe.
Different from sterilization methods such as high temperature and high pressure, there will be a certain amount of residue after ethylene oxide sterilization. The sterilized veteriary syringe needs to be placed for a period of time to release the residue and ensure the safety of dosing syringe and medicine.